Model 240 Hyperbaric Water Based Fire Extinguisher: Part Number Model-240 Includes
- (1) Wall Hanger
- Owner's Manual
- Water Capacity: 2.0 gal. (7.6 liters)
- Maximum Pressure: 165 psi (11.3 bar)
- Maximum Depth: 225 FSW
- Dimensions: 7.0 in. Dia. x 30 in. H (18 cm x 76 cm)
- Empty Weight: 6.50 lbs. (2.95 kg)
The Amron International Model 240 Hyperbaric Water Based Fire Extinguisher is a modified Class 2-A extinguisher and is tested to ANSI/UL711 and 626 standards and is designed for the pressures and confines of a hyperbaric chamber. The Model 240 has been fully tested and approved for use in all U.S. Navy standard double lock chambers.
In actual operation, the Amron International Model 240 Hyperbaric Water Based Fire Extinguisher is filled with potable water and then pressurized to 150 psig. The unit comes complete with a fill valve, pressure gauge and a pressure relief valve that is factory set at 173 psig to prevent over-pressurization. The Model 240 can be used in a chamber pressurized to a maximum depth of 225 FSW. If the unit is activated at the maximum depth, and held until the water is exhausted, the pressure inside the extinguisher will equal the chamber pressure. With the chamber and extinguisher at equilibrium, the fire extinguisher cannot implode or crush.
Brand Amron International Video No
Does the MODEL-240 Fire Extinguisher require any maintenance?
Yes, inspect monthly or at more frequent intervals when circumstances require. Make sure the hose and nozzle are unobstructed. Gauge pressure must be in operable range 0-200 PSIG. Lockpin must be in place. A visual inspection should be preformed annually and a hydrostatic test preformed every 5 years. Fire extinguisher laws vary in different regions of the world please check the laws that govern your area.
Has the Model-240 Fire Extinguisher been classified by ABS, Lloyds, DNV, Germanischer Lloyd, or any other classifying agency?
No. However, it was tested and approved for use by the U.S. Navy.
Where on the cylinder can I find the date the tank was originally hydrostatically tested?Per the manufacturer, the manufacture date (year only) is located on the lower right side of the large instructions label on the center of the tank. Your fire extinguisher should be hydrostatically tested every 5 years. Fire extinguisher laws vary in different regions of the world please check the laws that govern your area.
Shouldn’t there be a month and year of manufacture on my fire extinguisher’s name plate?No. The rules and regulations that govern this product only state there has to be a year of manufacture. When hydrostatically testing your tank you have until December 31 of the year the hydrostatic test is due to hydro your tank.
Customers Who Bought This Item Also Bought
List Price: $255.00$230.78